To describe a new technique for superficial implantation of the I-Stop TOMS transobturator sling and present the clinical outcomes on patients treated for mild to moderate urinary incontinence after radical prostatectomy.
We evaluated the four-arm sub urethral sling I-Stop TOMS performed in our institution between March 2012 and March 2015 using a superficial implantation technique. After a small incision, the perineal aponeurosis was incised but no muscle dissection was performed. Inclusion criteria for sling procedure used in the study was mild (1-2 pads/day) to moderate (3-5 pads/day) postprostatectomy incontinence and at least 12 months after radical prostatectomy. Improvement was defined as the patient having 50% reduction in the number of pads and success as patient either not wearing pads or using one security pad. The primary objective was to evaluate the proportion of patients achieving continence after the modified sling implantation technique at 12 months after surgery.
Fifty-two patients underwent our new technique and 34 had completed 12 months follow-up. The procedure was successful for 28 patients (82.4%). There was an improvement in 25 patients (73.5%). Pad use at 12 months had decreased significantly compared with baseline (mean 0.7 vs 2.2, P < .00001). The complications were rare and the procedure was well tolerated as shown by median visual analogic scale of 1.5 (interquartile range, 2 to 1).
This novel approach for insertion of the transobturator I-Stop TOMS male sling is a quick, simple, and well-tolerated procedure with low complication rate, allowing a significant improvement in postprostatectomy incontinence.
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Urology
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- EAU guidelines on urinary incontinence.Actas Urol Esp. 2011; 35: 373-388
- Post-prostatectomy urinary incontinence: a confluence of 3 factors.J Urol. 2010; 183: 871-877
- Surgical treatment of stress incontinence in men.Neurourol Urodyn. 2010; 29: 179-190
- Postprostatectomy incontinence: all about diagnosis and management.Eur Urol. 2009; 55: 322-333
- Contemporary management of postprostatectomy incontinence.Eur Urol. 2011; 59: 985-996
- Fourth International Consultation on Incontinence Recommendations of the International Scientific Committee: evaluation and treatment of urinary incontinence, pelvic organ prolapse, and fecal incontinence.Neurourol Urodyn. 2010; 29: 213-240
- Transobturator male sling TOMS for the treatment of stress post-prostatectomy incontinence, initial experience and results with one year's experience.Int Braz J Urol Off J Braz Soc Urol. 2009; 35 (discussion 714-5): 706-713
- Prospective evaluation of the functional sling suspension for male postprostatectomy stress urinary incontinence: results after 1 year.Eur Urol. 2009; 56: 928-933
- The AdVance transobturator male sling for postprostatectomy incontinence: clinical results of a prospective evaluation after a minimum follow-up of 6 months.Eur Urol. 2009; 56: 923-927
- I-STOP TOMS transobturator male sling, a minimally invasive treatment for post-prostatectomy incontinence: continence improvement and tolerability.Urology. 2012; 79: 458-463
- The two-year outcome of the I-Stop TOMS™ transobturator sling in the treatment of male stress urinary incontinence in a single centre and prediction of outcome.Prog En Urol J Assoc Fr Urol Société Fr Urol. 2013; 23: 1494-1499
- Postoperative male stress urinary incontinence: outcomes of treatment by I-STOP TOMS® transobturator sling.Prog En Urol J Assoc Fr Urol Société Fr Urol. 2014; 24: 127-131
- Update on 2-year outcomes of the TOMS™ transobturator male sling for the treatment of male stress urinary incontinence.Neurourol Urodyn. 2014; (In press)
- Comparison of clinical outcomes between “ideal” and “nonideal” transobturator male sling patients for treatment of postprostatectomy incontinence.Urology. 2014; 83: 1186-1188
- Mid-term evaluation of the transobturator male sling for post-prostatectomy incontinence: focus on prognostic factors.BJU Int. 2011; 108: 236-240
Published online: December 24, 2015
Accepted: October 23, 2015
Received: September 9, 2015
Marc Galiano and Cyrille Guillot-Tantay contributed equally.
Financial Disclosure: Marc Galiano is a consultant to American Medical Systems and CL Medical. The remaining authors declare that they have no relevant financial interests.
© 2016 Elsevier Inc. Published by Elsevier Inc. All rights reserved.